Here are some CT scan images of a 73 year old lady with a proximal humeral fracture
Her surgeon attempted a reconstruction with a massive cemented humeral implant. Note the greater tuberosity applied to the metal prosthesis laterally.
She presented to us a year later with pseudo paralysis and with resorption of the tuberosity as shown in the film below.
With difficulty in removing the cemented stem, we were able to convert to a reverse total shoulder.
We saw her yesterday in the office - she said she had no shoulder pain and had active elevation to 150 degrees. Her films are shown below.
Comment: Complex fractures such as this may be best managed by an initial reverse total shoulder rather than attempting an anatomic prosthesis. In cases where a prosthesis was previously cemented, a reverse can be cemented within the existing cement mantle.
You may be interested in some of our most visited web pages including:shoulder arthritis, total shoulder, ream and run, reverse total shoulder, CTA arthroplasty, and rotator cuff surgery as well as the 'ream and run essentials'
See from which cities our patients come.
See the countries from which our readers come on this post.